|Minimum Order Quantity||1 Piece|
|Output||4 to 20 mA.(non-isolated)|
capacitance is formed between the probe and the vessel wall.As the level of the measuring medium rises or falls capacitance between the probe and vessel wall changes proportionately. A high-frequency sine wave is applied to the probe and tank wall. A change of medium level will induce a change in the current of the applied sine wave. From when the capacitance in the vessel increase, the impedance will decrease and current will increase and vice-versa. (Z=Resistance, C=Capacitance, F= Frequency)
It must be ensured that the probe is parallel to the tank wall throughout its length.
kabir make Capacitance type 3 wire Level transmitter Model (CLT-1) with noninsulated probes for non-conductive liquids or PTFE Teflon insulated probes for conductive liquids.
|Frequency of operation||5 to15kHz.|
|Probe Material||Stainless Steel / Teflon insulated 10/12/16 mm rod up to Max 3 Metre length|
|Process connection||3/4",1",1.1/2" BSP(M) or 1 "ASA 150 lbs flanged but thickness 12mm (or as per customer specification)|
|Output ||4 to 20 mA.(non-isolated)|
|Span & Zero adjustment||Trim pots|
|Accuracy||±0.5%, ±1% or ±2%|
|Enclosure||Cast aluminium weather proof with PG9 cable gland or Flame proof (IS2148-2004,Gr II A and B gases)|
|Cable Size||3 core, 0.75 sq.mm screened PVC cable.|
While sending enquiries for capacitance type level transmitters, please furnish following informations to enable us to submit a suitable offer: